Digital Coating Thickness Gauge is excellent for measuring coating thickness on metal surfaces.
Display: 4 DIGIT LCD
Minimum curvature radius: protruding 1.5 mm , concave 6mm
Resolution: 1 micrometre
Measurement Rate: 0.8 seconds
Accuracy: +/-(3% + 2 micro meter)
Range: 0-1200 micrometre
Critical thickness of metallic base: 1 mm Minimum thickness of metallic base: 0.2
Minimum Display Unit: micrometer or mil
Probe diameter: 13 mm
Frequently asked Question
What is coating thickness gauze?
Coating Thickness Gauge is such coating thickness in Type as well as displaying as paint, plus this gadget measures digital way to indicate the depth of coating both F gauge is to the LCD Display be used to measure dry film results on This and NF the non-destructive way
How to measure Dry Film Thickness?
Dry film thickness (DFT) can be a coating or the measurement, using techniques substrate using a substrate such as magnetic thickness measurement, where the coating measured using two methods: destructive is cut to the coating thickness a cutter; and non-destructive which do not damage the magnetic induction and eddy current thickness measurement methods.
You can also check out our Machinery Products.